Home - Education Resources - NDT Course Material - EC Testing
 - Eddy Current Testing Introduction Basic Principles History of ET Present State of ET The Physics Properties of Electricity Current Flow & Ohm's Law Induction & Inductance Self Inductance Mutual Inductance Circuits & Phase Impedance Depth & Current Density Phase Lag Instrumentation Eddy Current Instruments Resonant Circuits Bridges Impedance Plane Display - Analog Meter Probes (Coils) Probes - Mode of Operation Probes - Configuration Probes - Shielding Coil Design Impedance Matching Procedures Issues Reference Standards Signal Filtering Applications Surface Breaking Cracks SBC using Sliding Probes Tube Inspection Conductivity Heat Treat Verification Thickness of Thin Mat'ls Thickness of Coatings Advanced Techniques Scanning Multi-Frequency Tech. Swept Frequency Tech. Pulsed ET Tech. Background Pulsed ET Remote Field Tech. Quizzes Formulae& Tables EC Standards & Methods EC Material Properties -

Resonant Circuits

Eddy current probes typically have a frequency or a range of frequencies that they are designed to operated.  When the probe is operated outside of this range, problems with the data can occur.  When a probe is operated at too high of a frequency, resonance can occurs in the circuit. In a parallel circuit with resistance (R), inductance (XL) and capacitance (XC), as the frequency increases XL decreases and XC increase.  Resonance occurs when XL and XC are equal but opposite in strength.  At the resonant frequency, the total impedance of the circuit appears to come only from resistance since XL and XC cancel out.    Every circuit containing capacitance and inductance has a resonant frequency that is inversely proportional to the square root of the product of the capacitance and inductance.

In eddy current probes and cables, it is commonly stated that capacitance is negligible.  However, even circuits not containing discreet components for resistance, capacitance, and inductance can still exhibit their effects.  When two conductors are placed side by side, there is always some capacitance between them.  Thus, when many turns of wire are placed close together in a coil, a certain amount of stray capacitance is produced.  Additionally, the cable used to interconnect pieces of electronic equipment or equipment to probes, often has some capacitance, as well as, inductance. This stray capacitance is usually very small and in most cases has no significant effect. However, they are not negligible in sensitive circuits and at high frequencies they become quite important.

The applet below represents an eddy current probe with a default resonant frequency of about 1.0 kHz. An ideal probe might contain just the inductance, but a realistic probe has some resistance and some capacitance. The applet initially shows a single cycle of the 1.0 kHz current passing through the inductor.

Exercise 1: Using your mouse, adjust the resistance by sliding the slide bar. Does the frequency change?

Exercise 2: Note that changing the inductance and/or the capacitance changes the resonant frequency of this resonant circuit. Can you find several combinations of capacitance and inductance that resonate at 1.0 kHz?